# Measurement of Low value resistance (0.1 to 50 Ohm)

The previous voltage divider design is suitable for only measurement of higher values of resistance (50 Ohm to 1 MOhm) to measure very small value resistance we have built this circuit, it consists of constant current source of 100mAmp. Measuring voltage drop across a resistor having constant current gives us resistance value (R = V / I).
Calculations
V = I x R
R = V / I
Where:
I = 100mAmp as we are using constant current source of 100mAmp
V = Measured by Arduino
Constant current source is built using LM317
Iconst = Vref / R where: Vref of LM317 is 1.25V
Iconst = 1.25 / R
Iconst = 104mAmp

## Result of Resistance Measurement

Open the serial terminal and see the resistance value. This code and circuit will not work for values more than 50 Ohm as 50 * 100mA = greater than 5V.

## 2 thoughts on “Measurement of Low value resistance (0.1 to 50 Ohm)”

1. Malosse says:

Hello

Is it possible to reduce constant courant source to 10mA ?

Manu

1. When we reduce current voltage drop across resistance becomes small, you can reduce it.. Use differential amplifier to amplify voltage across resistor. Read more on bridge circuit for resistance measurement.